- Revision Hip Resurfacing Patients Have Better Outcomes Than THR Revision Patients
- Loose or Slipped Acetabulum Cups by Dr. Bose
- DePuy Orthopaedics Discontinuation of ULTAMET Metal-on-Metal
- Dr. Su responds to recent Lancvet Article about failoure rates
- Knowing the cause of resurfacing failure can ensure successful conversion to THR by Dr. Su
- I Have Bone Cysts, Can I Have a Hip Resurfacing?
- Bone Cysts by Dr. Scott Rubinstein
- Pseudotumours Risk For Hip Resurfacing Highlights
- DuPuy Reaches
June 1, 2018 – BARCELONA, Spain – Total surface replacement may still be a viable hip arthroplasty option in young, active patients based on how well it fared in a study when it was revised to total hip replacement vs. primary THR that was revised to THR, according to a presenter at the EFORT Annual Congress…
…“Our study shows, therefore, that there is less blood loss, there’s a shorter length of … Read the rest
The issue of cup slippage in the immediate postop period is a controversial one.
While bone ingrowth takes around 6 wks. – the hydroxy apatite to bone chemical reaction can occur much more quickly.
If we surgeons feel that the cup is not perfectly tight ( press fit) during the surgery then we restrict activities for a 6 -8 wk period .This is done in the hope that no precipitating event would occur that would … Read the rest
DePuy Orthopaedics, Inc. Statement on
Discontinuation of ULTAMET Metal-on-Metal and
COMPLETE Ceramic-on-Metal Hip Systems
DePuy Orthopaedics, Inc., announced its decision
to discontinue sales of its ULTAMET
Metal-on-Metal Articulation and COMPLETE
Ceramic-on-Metal Acetabular Hip System
worldwide. The discontinuation will be effective
August 31, 2013. This will allow surgeons to
plan accordingly for upcoming surgeries. The
ceramic head used in COMPLETE will continue to
be available for use in other bearing surface
The metal liner that … Read the rest
Read Complete Article by Clicking Here
I would like to take this opportunity to comment on the Lancet article, “Failure rates of metal-on-metal hip resurfacings: analysis of data from the National Joint Registry for England and Wales”, by Professor AW Blom, published on October 2, 2012.First of all, this is an observational scientific study with valid research design and questions; however, the conclusions point out the limitations of registry studies (more on this to
Knowing the cause of resurfacing failure can ensure successful conversion to THR by Edwin Su, MD
The shell can be retained in cases involving femoral neck fracture, femoral loosening or impingement.
Causes of failure
“The cause of failure must be carefully assessed prior to the conversion surgery in order to ensure an optimal THR outcome,” Su said. He noted that femoral neck fracture is the primary cause of short-term failure in resurfacing procedures.
Dr. Bose – I have bone cysts, can I have a hip resurfacing?
The presence of a cysts by itself is not a contraindication for resurfacing. It does not preclude resurfacing automatically. One must keep in mind that cyst formation is a natural occurrence in osteoarthritis and is very common though the extent, quantity & location may vary.
Cysts are of course much more common and invariably present in AVN. The assessment of certain technical … Read the rest
|By Dr. Rubinstein|
2860 N. Broadway Suite 202
Chicago, IL 60657
|Cysts are very common in arthritic hips and can usually be|
seen on x-rays. Most of the time they are not a problem with
resurfacing because they are small and in the part of the bone removed in
milling the femoral head for a resurf. Most larger ones can be filled
with a cement of bone chips and still do
July 11, 2008As the incidence of metal-on-metal hip resurfacing has increased in recent years, especially in younger patients, research published in the Journal of Bone and Joint Surgery – British Volume (JBJS-Br) discusses occurrences of “pseudotumours” as a result.The researchers estimate that approximately 1% of patients who have metal-on-metal hip resurfacings develop pseudotumours within five years of treatment. A pseudotumour is ‘a soft-tissue mass associated with the implant…neither malignant nor infective
If an HRA has failed, I try to solve the problem by revising only the acetabular component whenever possible because HRA are functionally better than THR. If only the acetabular component is revised, the patient still has a HRA. If the femoral component is revised, the patient now has a THA.
Currently revising only the acetabular component with a metal/metal (M/M) articulation is only … Read the rest
By Dr. Koen De Smet & Dr. Alessandro Calistri
Acoustic Phenomena in Hip Resurfacing 2009 Koen De Smet, MD; Alessandro Calistri, MDAMC Ghent, Krijgslaan 181, 9000 Ghent, BelgiumBackground: While providing superior hardness and improved wear characteristics, hard bearings such as metal-on-metal, or ceramic-on-ceramics bearings have different lubrication properties than hard on soft (metal- or ceramic-on-poly) bearing couples. A failure in this lubrication mechanism in metal-on-metal resurfacing prostheses can lead to squeaking… Read the rest
Memo sent from Dr. Dorr to the American Association of Hip and Knee Surgeons that deals with what he feels is the cause for the hip implant failures:MEMO DATE: 4/22/08 TO: American Association of Hip and Knee Surgeons
FROM: Larry Dorr, M.D.
RE: This NOTICE is to inform you that we have had ten revisions in 165 hips and have four more that need to be revised using the Durom cup (Zimmer, Inc)This
Javad Parvizi, MD, FRCS, Michael Leunig, MD and Reinhold Ganz, MD
Dr. Parvizi is Associate Professor, Rothman Institute at Thomas Jefferson University Hospital, Philadelphia, PA. Dr. Leunig is Associate Professor, Department of Orthopedic Surgery, Balgarist University Hospital, Zurich, Switzerland. Dr. Ganz is Professor, Department of Orthopedic Surgery, Balgarist University Hospital.None of the following authors or the departments with which they are affiliated has received anything of value from or owns stock in a commercial
Revision of metal-on-metal resurfacing arthroplasty of the hip – THE INFLUENCE OF MALPOSITIONING OF THE COMPONENTS 2008
R. De Haan, MD, Resident in Orthopaedic Surgery1; P. A. Campbell, PhD, Research Scientist2; E. P. Su, MD, Orthopaedic Surgeon3; and K. A. De Smet, MD, Orthopaedic Surgeon4 1 University Hospital Brussels, Laarbeeklaan 101, 1090 Brussels, Belgium. 2 J. Vernon Luck Snr MD, Orthopaedic Research Centre, Orthopaedic Hospital/University of California Los Angeles, 2400 South Flower Street, Los Angeles, … Read the rest
During my several years of learning and reading about hip resurfacing, I have heard of two people having dislocations. There may be more, but I only know if people post their dislocation on the various discussion boards I belong to.
One young man was doing an exercise bringing his knee to his chest. His leg was tight and he jerk his knee toward his chest and caused a dislocation of his hip resurfacing.
One lady … Read the rest
The issue of cup slippage in the immediate postop
period is a controversial one.
While bone ingrowth takes around 6 wks. – the
hydroxy apatite to bone chemical reaction can occur
much more quickly.
If we surgeons feel that the
cup is not perfectly tight ( press fit) during the
surgery then we restrict activities for a 6 -8 wk
period .This is done in the hope that no
precipitating event would occur that would … Read the rest